AX Series (AX-400, AX-500, AX-600, AX700, AX-800, AX-900) Anesthesia Machine offers SIMV supports the smooth transition to anesthesia resuscitation, promotes resuscitation efficiency, and reduces lung injury caused by man-machine confrontation. CPAP/PSV supports postoperative respiratory therapy to help patients to wean successfully and reduce the dose of sedatives and muscle relaxants applied.
PCV, PS+ Apnea ventilation modes provide more flexibility to ventilation strategy, suitable for various patients' operations including pediatric and adult.
Advanced ventilation management, intelligent switch for key parameters in the conversion process of ventilation mode.
Support both low-flow anesthesia and min-flow anesthesia.
Electronic flow-meter, precise control, and convenient operation.
Intraoperative anesthetic gas usage statistics: easy for medical staff to know the cumulative dose of medication
Statistics of soda-lime consumption: early warning of the consumption of soda lime to prevent CO2 retention
Ventilator risk operation reminder: misoperation warning, risk operation prompt check, reduce operation accident.
Shutdown delay: avoid accidental ventilation stop caused by misoperation.
The ascending bellows pneumatic electronic control technology provides a more compact circuit and a smaller loop cavity, which realizes accurate ventilation.
ANESTHESIA GAS SUPPLY MODULE
O2, N2O, AIR; 280j600kPa
Optional: O2, N2O, AIR
Fresh gas flow indicator
Electronically display and virtual fresh gas flow tubes
Range of fresh gas flow indicators
O2, N2O (0~10.0 L/min), AIR (0~12.0L/min) (increments of 0.05)
Auxiliary common gas outlet (ACGO)
Anesthetic Gas Scavenging System (AGSS)
Sevoflurane, Halothane, Enflurane, Isoflurane
VENTILATOR OPERATING SPECIFICATIONS
Pneumatically driven, Electronically controlled
Ventilator Monitoring & Alarms
l Volume control (IPPV)
l Pressure control (PCV)
l Pressure Controlled Ventilation Volume Guaranteed (PCV-VG)
l Synchronized Intermittent Mandatory Ventilation in Volume (SIMV-VC)
l Synchronized Intermittent Mandatory Ventilation in Pressure (SIMV-PC)
l Pressure support (PS) with Apnea Backup
Breathing frequency (Freq)
Positive end-expiratory pressure (PEEP)
OFF, 3~30 cmH2O
Inspiration/expiration ratio (I:E)
Tidal volume (Vt)
20~1500 ml in Volume Control